The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Agilent A. 20 MHz Function /. Arbitrary Waveform Generator. Service Guide . Publication Number (order as manual set). Agilent A. Function/Arbitrary Waveform Generator .. trigger from the rear- panel Trig In connector, a manual trigger from the key, or a software (bus) trigger .
|Published (Last):||8 August 2009|
|PDF File Size:||16.65 Mb|
|ePub File Size:||9.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page Chapter 3 Features and Functions Frequency Sweep Start Frequency and Stop Frequency The start frequency and stop frequency set the upper and lower frequency bounds for the sweep. Index Index If you have questions relating to the operation of the Agilent A, call in the United States, or contact your nearest Agilent Technologies Office.
Chapter 5 Error Messages Execution Errors Settings conflict; FSK turned off by selection of other mode or modulation The function generator will allow only one modulation, sweep, or burst mode to be enabled at the same time.
This error is also reported when a character-type parameter is too long. The variation in phase of the modulated waveform from the carrier waveform is called the phase deviation, which can vary from 0 to degrees. You can output the burst at a rate determined by the internal rate generator or the signal level on the rear-panel Trig In connector. The default is Internal. You can also select an infinite burst count. OFFS commands to configure the carrier waveform.
Chapter 5 Error Messages Execution Errors Settings conflict; edge time decreased due to pulse width For a pulse waveform, the function generator will automatically adjust the waveform parameters in the following order as needed to generate a valid pulse: This rear-panel connector is used in the following modes: Page Chapter 5 Error Messages Arbitrary Waveform Errors Not able to delete a built-in arb waveform You cannot delete any of the five built-in waveforms: The default is 1 ms.
This is a read-only register and bits are not cleared when you read the register. You can name a location from the front panel or over the remote interface, but you can only recall a state by name from the front panel. Burst Mode Overview The following is an overview of the steps required to generate a burst.
The default is OFF. You cannot use pulse, noise, or dc as the carrier waveform.
33220A Function / Arbitrary Waveform Generator, 20 MHz
This section describes the low-level commands 33202a to program the function generator to output a pulse waveform. Chapter 3 Features and Functions Burst Mode Burst Period The burst period defines time from the start of one burst to the start of the next burst.
Page of Go. Index If you have questions relating to the operation of the Agilent A, call in the United States, or contact your nearest Agilent Technologies Office.
Agilent Technologies 33220A User Manual
Index FREQuency command center frequencyfront-panel operation 38 FUNCtion command external trigger source FSK rate 39, 91 modulating frequency 82, frequency spanhop frequency 38, 91, modulating source 84, front-panel operation 40, 42 modulating source 92, modulating waveform 82 linear vs. Each character must be a letter “a” IP address to a device on a network. Page Chapter 5 Error Messages Execution Errors Settings conflict; FSK turned off by selection of other mode or modulation The function generator will allow only one modulation, sweep, or burst mode to be enabled at the same time.
Square waveforms are created by routing a DDS-generated sine wave into a comparator. Chapter 5 Error Messages Command Errors Program mnemonic too long A command header was received which contained more than the maximum 12 characters allowed. Check the list of parameters to verify that you have used a valid parameter type.
Chapter 3 Features and Functions System-Related Operations Display Control For security reasons, or to speed up the rate at which the function generator can execute commands from the remote interface, you may want to turn off the front-panel display. You can select one of the five built-in arbitrary waveforms, one of four user-defined waveforms, or the waveform currently downloaded to volatile memory.
Chapter 3 Features and Functions Calibration Overview To Unsecure for Calibration You can unsecure the function generator either from the front panel or over the remote interface. Select the source from which the function generator will accept a trigger.
Press the IP Setup softkey. When enabled using the OUTP: The GPIB address can take any value between 0 and Chapter 5 Error Messages Arbitrary Waveform Errors Not able to delete a built-in arb waveform You cannot delete any of the five built-in waveforms: Chapter 3 Features and Functions Frequency Modulation FM Frequency Deviation The frequency deviation setting represents the peak variation in frequency of the modulated waveform from the carrier frequency. Burst Chapter 7 Tutorial Burst connector can be configured as an output to enable the A to trigger other instruments at the same time as its internal trigger occurs.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Page 97 Chapter 3 Features and Functions Pulse Width Modulation PWM Width Deviation The width deviation represents the variation in width in seconds in the modulated waveform from the width of the original pulse waveform.
Page 8 In This Book Quick Start Chapter 1 prepares the function generator for use and helps you get familiar with a few of its front-panel features.
The magnitude of a sine manuwl is usually specified by its peak, peak-to-peak, or root-mean- square RMS value. The function generator will accept an internal or external modulation source.
This error may be the result of a device failure or extreme conditions such as lightning or strong magnetic fields. This chapter describes basic signal-generation concepts and gives specific details on the internal operations of the function generator. To configure the function generator to recall the factory default state at power-on, press and then select the Pwr-On Default softkey. Cross-isolation UART framing error To set the waveform period instead, press the Freq softkey again to toggle to the Period softkey.
The rear-panel Trig connector cannot be used for both operations at the same time. Used only when the Internal modulation source is selected PWM: Page 65 Chapter 3 Features and Functions Output Configuration Duty Cycle Square Waves The duty cycle of a square wave represents the agolent of time per cycle that the square wave is at a high level assuming that the waveform is not inverted.
MIN selects the most negative dc offset voltage for the selected function and amplitude. At power-on, the units for output amplitude are volts peak-to-peak.
Instrument State Storage The function generator has five storage locations in non-volatile memory to store instrument states. Be sure to use mmanual support rails in the rack cabinet. The CD auto-starts and provides a Welcome page with instructions. To set a password use the knob and cursor keys use to delete all characters to the right of the cursor position.